Although the use of pulmonary artery catheters (PACs, otherwise known as Swan-Ganz catheters) has never been shown to confer a mortality benefit, they are still routinely used in operating rooms and intensive care units to help guide resuscitation, monitor pulmonary artery pressures (PAP), differentiate mixed shock states, assess continuous cardiac output (CCO), etc. In this post, I’ll be discussing the anatomy of an Edwards Swan-Ganz CCOmbo/VIP PAC. Keep in mind that PACs come in many flavors, so depending on the exact catheter, the number of ports, length, etc. will vary.
PA CATHETER STRUCTURE
This PAC is a quadruple lumen catheter that measures core temperature, CCO, central venous pressure (CVP), and PAP. Additionally, one can obtain a left ventricular end diastolic pressure (LVEDP) and true mixed venous blood sample.
Each thin dash represents 10 cm and the thick dash represents the 50 cm mark. This PA catheter is 110 cm long and includes the following:
- PA port (distal tip, 0 cm) with a balloon. Used to transduce PA pressures.
- Syringe to fill 0.5 – 1.5 cc of air to inflate the balloon. There will be a locking mechanism to keep the balloon inflated or deflated.
- CVP port (30 cm from the tip). This port will reside in the right atrium or cavoatrial junction. Useful to transduce central venous pressure (CVP)
- Proximal injectate/variable infusion port (~31 cm from the tip). Usually in the cavoatrial junction or superior vena cava and useful to infuse vasoactive agents.
- Cable connector to monitor cardiac output on a compatible Edwards device.
Before floating the PA catheter, flush all the ports and verify that the balloon inflates symmetrically and collapses completely. Also ensure that the pressure transducers are zeroed appropriately.
FLOATING THE SWAN
Placing the PAC requires a central vein introducer. Once the PAC is advanced past the lumen of the introducer (~10-15 cm), the balloon is inflated and “floats” to the pulmonary artery following the normal path of blood flow: right atrium → tricuspid valve → right ventricle → pulmonic valve → pulmonary artery. By transducing the waveform on the tip of the PAC, we can tell which chamber of the heart the catheter resides in. In cardiac cases where I’ll be using intraoperative transesophageal echocardiography (TEE), I usually put the TEE in first and use it to help guide my passage of the PAC in addition to the waveform.
While advancing the catheter, one must be aware of any arrhythmias, however transient, as these can cause the patient to collapse. A patient with critical aortic stenosis, for example, relies on an atrial kick (during sinus rhythm) to augment their cardiac output. With an arrhythmia, the atrial kick is lost and cardiac output can fall precipitously leading to myocardial (and systemic) hypoperfusion, decreased contractility, even lower cardiac output… even worse hypoperfusion… yeah, downward spiral = BAD.
For a right internal jugular entry, typically I advance the PAC 40-50 cm in an adult before entering the pulmonary artery. At this point, I’ll lock a sterile sheath over the PAC and onto the introducer. Always remember to deflate the balloon. If concerned about coiling within a chamber, also remember the deflate the balloon before withdrawing the catheter and trying to re-float it.